|Published (Last):||13 April 2010|
|PDF File Size:||9.4 Mb|
|ePub File Size:||7.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
EDAN owns the copyrights of this manual. Without prior written consent of EDAN, any materials contained in this manual shall not be photocopied, reproduced or translated into other languages. Page 3 Terms Used in this Manual This guide is designed to give key concepts on safety precautions. Page 5 5. Therefore, your model may or may not have some of the parameters and functions described, depending on what you have ordered.
If the damage is evident, replace the defective part s before use. Type B applied part. The symbol indicates that the device should be sent to the special agencies according to local regulation for separate collection after its useful life. Take 2. When the battery is low, this symbol flashes; the Doppler will shut down automatically after 7 seconds.
Page Buttons Doppler. The 2MHz obstetrical probe features in deep penetration and is designed for use during the third trimester pregnancy. MS Part number of 4. MS3R Part number of 5. Page Batteries Do not stretch the probe cable for more than two meters long.
Figure b Figure Connecting battery connector 3 Close the compartment. Slide the cover back to close the battery compartment. Pinch the probe head and pull it outwards with mild force, and then take out the whole probe slantwise in the direction of the arrow. Figure Taking out the probe 2 Placing the probe Hold the main unit with one hand.
Sonotrax Basic A Fetal Doppler Baby Heart Monitor
Doppler vascular Edan SonoTrax
EDAN SonoTrax Vascular Doppler 8Mhz Review